Payment & Shipping Terms:
|Aplication:||Salt Spray Machine||Warranty:||12 Months|
|Usage Strength:||Test Chamber||Power:||1φ、220V/50HZ|
|Spray Capacity:||1~10cc/h||Structure Material:||F.R.P|
|Standard:||ISO-9227 ASTM-B117 JIS-D0201 H8502 H8610 K5400 Z2371||Internal Size:||90×60×50cm|
270L Salt Mist Test Chamber For Alkaline Corrosive Test
This machine for all kinds of metal material surface treatment, including coating, electroplating, organic and inorganic coating, anodizing, rust proof oil and preservative treatment test of its corrosion resistance, so as to establish the quality of the products.
|Inner box size:
|Outside box size:
|Equipment material:||Both internal and external material is using imported P.V.C rigid plastic board; the case cover is using transparent rigid plastic board P.V.C.|
|Test chamber temperature||NSS ACSS 35±1°C CASS:50±1°C|
|Saturated air barrel temperature:||NSS ACSS 47±1°C CASS 63±1°C|
|Spray quantity:||1.0~2.0 ml / 80cm2 /hr|
|PH:||NSS ACSS6.5~7.2 CASS 3.0~3.2|
|Brine tank capacity:||15L||25L||40L|
|Multiple safety protection devices:||Water shortage/ over temperature / phase protection|
|Standard accessories:||Test salt, measure bucket, tool etc 1 set|
Detailed configurations as follows:
1, machine material:
(A) testing room using PVC polyvinyl chloride plate (from Taiwan), acid-resisting and alkali-resisting, cleaning easier.
(B) cover of testing room using acrylic panels (imported organic glass), the thickness is 5mm.
(C) reagents added bottles attached with water gauge, cleaning easy and not easy to break.
(D) saturated air barrels is made of SUS # 304 stainless steel.
(E) testing room to adopt indexing frame, can be arbitrarily adjust the angle, Surrounded by fog and placed more piece .
2, According to CNS JIS ASTM GB2423.19 specifications, can be set constant temperature to control
(A) Salt spray test: NSS, ACSS
1. testing room : 35 ℃ ± 1 ℃
2 saturated air barrels : 47 ℃ ± 1 ℃
(B) anti-corrosion test: CASS (copper acetate)
1.testing room: 50 ℃ ± 1 ℃
2 saturated air barrels: 63 ℃ ± 1 ℃
3, Air supply system: air pressure 1Kg/cm2 Two-stage adjustments (coarse / fine-tuning).
(A) the first paragraph for the rough adjustment 2-2.5Kg/cm2, the second paragraph to fine-tune 1Kg/cm2.
(B) Testing configuration is switch connector designed for SO2.
4, Spray form:
(A) Using Bo Nute principle to draw brine and then saline spray, atomization uniform, non-blocking and crystallization, so that ensure continuous spraying.
(B) Nozzle: The nozzle is made of special glass, adjust spray volume and spray angle.
(C) spray volume: Adjustable 1 ~ 2ml (80cm2/hr) (average of 16-hour volume).
5, Heating devices:
By direct heating, warming up faster and reduction of standby time, when the temperature arrives automatically switch ON / OFF, make temperature accuracy, low power consumption (heating tube is made of titanium alloy).
6, Control system:
(A) testing room temperature controller is the digital micro-computer products 0-99.9 ℃, liquid crystal digital display with PID automatic calculation, the control error is ± 0.3 ℃ (YANG MING6802) 1 pcs.
(B) saturated air barrels controller 0 ~ 99.9 ℃, liquid crystal digital display with PID automatic calculation, the control error is ± 0.3 ℃ (YANG MING6802) 1 pcs.
(C) safe temperature controller of heat expansion of the liquid tank 30 ~ 110 ℃ (Italy) 1 pcs.
(D) safe temperature controller of expansion of the saturated air barrel : 30 ~ 110 ℃ (Italy) 1 pcs.
(E) digital time controller 9.9s ~ 9990hr (Taiwan) 1 pcs.
(F) timer 0 ~ 9999hr (Taiwan Anliang) 1 pcs.
(G) buckle switch, continuous buckling 20000 times.
(H) 60 type, 90 type for manual flip, 120, 160, 200 type for the pneumatic cylinde automatic opening and closing.
(J) pressure gauge, (Taiwan B card).
(K) fixture bleaker (YANG MING).
(L) electromagnetic valve; (KSD).
(M) regulating valve (YIHUA).
(N) bleaker (Omron).
7, Add water systems:
Using automatic water replenishment system, automatic replenishment when the water level is too low.
8, Safe protective device:
(A) low water level, automatically cut off power supply.
(B) over-temperature, automatic cut off the power supply.
(C) attached to the safe warning light device.
Reagent sodium chloride (NACL) 500 g 2 bottles
Calibration cup 1 pcs
instruction 1 pcs
PH test paper 1 pcs